Introduction: 50% of critically ill patients fail to reach caloric targets with NG feeding. PP feeding may enhance caloric intake. PP feeding can be continued throughout theatre in patients with a secure airway. Blind PP tube placement is difficult. CEAS has been developed to assist tube placement and eliminate check X-rays of tube position. Method: All BITU patients with CEAS PP feeding tube placement were identified. Notes and X-rays were reviewed. Tube position, calorie deficit and time off feed were recorded. Results: 44 tubes were placed in 21 patients using CEAS. 84% were PP, 16% NG. Position correlated to X-ray findings in 86%. In 16% position was NG on CEAS but was PP on X-ray. 10 patients required both CXR and AXR to confirm position, the remainder required CXR only. Time off feed varied from 0-24 h (mean 7.4 h). Calorie deficit ranged from 0-2465 kCal (mean 858 kCal). Average wait for X-ray was 3.4 h. If X-ray wait was eliminated calorie deficit would be reduced by 45% to 393 kCal. Conclusion: The Cortrak system is safe and effective on BITU. It reduces calorie deficit, reduces X-ray exposure and is cost effective. We recommend its use on BITU.
- Cortrak System™
- Post-pyloric feeding
ASJC Scopus subject areas
- Emergency Medicine
- Critical Care and Intensive Care Medicine